80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Published (Last):||23 April 2017|
|PDF File Size:||9.70 Mb|
|ePub File Size:||4.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fluke 80T-IR / 80T-IR/E used or new for sale at used-line
Record the reading 80t-it on the DMM. Replace the battery if necessary. The foregoing information relates to product sold on, or after, the date shown below.
Verification Points Model Blackbody Temperature at 0. Save this item to a new parts list. Power up the UUT and allow it to stabilize for five minutes.
Accuracy figures are valid for a period of one year. You have chosen to save the following item to a parts list:.
Search the history of over billion web pages on the Internet. The product detailed below complies with the specifications published by RS Components. If your instrument fails to pass any fluks specifications, return it to the Eluke Eactory or Service Center for repair.
Save to parts list Save to parts list. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody 80y-ir.
Showing Results for Fluke 80T-IR / 80T-IR/E
In between measurements move the UUT to an environment not affected by the heat source. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.
Refer to Figure 1 for equipment configuration. Add to a parts list.
80T-IR TEMP PROBE R
Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. The Manufacturers disclaim all warranties including implied warranties of merchantability or 80t-iir for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website.
Save to an existing parts 80t-ur Save to a new parts list. May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i. Please select an existing parts list.
Fluke Infrared Temperature Probe M# 80t-ir | eBay
You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice. Adjust tluke setpoint of the Blackbody to output each of the temperatures in Table 2 at 0. Sending feedback, please wait Please enter a message.
Set the DMM to the 3. Thank you for your feedback. Repeat steps 1 through 5 for the remaining verification points called out in Table 2.
Reactivate by switching the UUT off and on again. The Manufacturers reserve the right to change this Information at any time without notice. With the UUT powered on, flume that the battery voltage is at least 7.