The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||10 October 2008|
|PDF File Size:||13.10 Mb|
|ePub File Size:||2.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter the existing password from Entry keypad and press Enter3. Theoutput 6632v to zero, CV turned off, and Protturned on. It also made me less worried that there were secret vermin in my apartment that were taking refuge in my bench. Or at least you can get close. System KeysThis is the blue, unlabeled key, which is also shown as Shift in this guide. Enter the new password from the keypad. Seechapter 5 for more details. It is designed to simplify the testingof digital cellular and mobile phones.
UnrThe output is unregulated. East Dane Designer Men’s Fashion.
Page 43 Front Panel Operation – 52 – Querying and Clearing Output ProtectionThe dc source will disable its output if it detects an overvoltage or overcurrent fault condition. Amazon Inspire Digital Educational Resources.
Press Shift, OV,Enter8. Disconnect all loads from the dc source. In either case, thedisplay may show an error message that indicates the reason for the failure.
Measurement samples may notcoincide with hpp actual maximumor minimum point of the waveform. When set to Fast mode, this switch disconnects the output capacitor that is located inside theunit. This chapter provides a preliminary 6362b to the dc source front panel. Page 24 3 – InstallationVoltage DropsThe load wires must also be large enough to avoid excessive voltage drops due to the impedance ofthe wires. It still seems to be in cal from 5 years ago.
Dc source status registers allow remote monitoring of a wide variety of dcsource operating conditions. Page 18 2 — General InformationSafety ConsiderationsThis dc source is a Safety Class 1 instrument, which means it has a protective earth terminal.
Your manual failed to upload Save this state to location 1. Who are these people? Hope yours treats you well! Thatterminal must be connected to earth ground through a power source equipped with a groundreceptacle. In general, if the wires are heavy enough to carry the maximum short circuit currentwithout overheating, excessive voltage drops will not be a problem.
Set the output current to its full scale value see Table B Save an operating state in location 1 as follows: Unless otherwise noted, specifications apply when measured atthe rear terminals after a minute hl period.
HP Agilent Keysight, 6632B
Readings within high currentlimits see tables B-3,4, 5 or 6. I used a similar Philips Power supply source before PEbut this thing is stuffed with only imprecise analog meters and is 6362b comparable with this fine HP voltage supply, and one does have to use an additional precise voltage meter I do use a Fluke A Multimeter, a quite old, but freshly calibrated, still precise voltmeter. Page 38 5 – Front Panel OperationFunction KeysRefer to the examples later in this chapter for more details on the use of these keys.
6632v for the DMM readingto stabilize.
– Photography – HP B
Refer to Table to calculate the voltage dropfor some commonly used AWG copper wire. Add to Wish List. Please make sure that you are posting in the form of a question. After much swearing, compressed air, and windex, the inside of the chassis looked like this:.
After much swearing, compressed air, and windex, the inside of the chassis looked like this: